Block diagram of the developed system. IA instrumentation amplifier, LPF low-pass filter, HPF high-pass filter, MUX multiplexer, and PC personal computer
The transient signals were used to detect the subject’s movement in
the bed (Figs. 2 , 3 ). .
Viewing this image requires a subscription. If you are a subscriber, please log in.
This image is from the article titled "Non-constraining sleep/wake monitoring system using bed actigraphy"
(from Medical & Biological Engineering & Computing), which is copyrighted by International Federation for Medical and Biological Engineering. For more information on the
copyright for this image, please refer to the full image caption and to the
The image is being made available for non-commercial purposes for subscribers to SpringerImages. For more information on what you are allowed to do with this image, please see our copyright policy.
To request permissions to use any copyrighted material, please visit the source document.
Report a copyright concern regarding this image.
Log in or register to save your favorite images and download them as high-quality PowerPoint or PDF files.
Log in or register to save your search criteria.
© Springer, part of Springer Science+Business Media.
Remote Address: 22.214.171.124 Server: 21